Physics 



Chemistry 



Traceability to the SI through the BIPM 



Related links 



Contact us 





Calibration and Measurement Capabilities Electricity and Magnetism 

In the CMCs uncertainty statements, the notation Q[a, b] stands for the rootsumsquare of the terms between brackets: Q[a, b] = [a^{2} + b^{2}]^{1/2} 

Result of the search 

Your selection : Electricity and Magnetism

 Georgia, GEOSTM (Georgian National Agency for Standards and Metrology) 
 Complete CMCs in Electricity and Magnetism for Georgia (.PDF file)   DC resistance standards and sources, high values. Resistance standards, resistance boxes, > 1.00E+06 to 2.00E+09   Absolute expanded uncertainty (k = 2, level of confidence 95%) in : 20E06R to 1.6E03R, with resistance R   Direct measurement with a DMM   Temperature: (23 ± 3) °C   Approved on 19 May 2015   Internal NMI service identifier: GEOSTM/EM21 to EM24  
Uncertainty table

 DC resistance standards and sources, intermediate values. Resistance standards, 10 k   Absolute expanded uncertainty (k = 2, level of confidence 95%) in : 5E06R, with resistance R   Substitution with a DMM   Temperature: (23 ± 3) °C   Approved on 19 May 2015   Internal NMI service identifier: GEOSTM/EM14 
 DC resistance standards and sources, intermediate values. Resistance standards, resistance boxes, > 1 to 1.00E+06   Absolute expanded uncertainty (k = 2, level of confidence 95%) in : (15E06R + 10E06) to 20E06R, with resistance R   Direct measurement with a DMM   Temperature: (23 ± 3) °C   Approved on 19 May 2015   Internal NMI service identifier: GEOSTM/EM14 to EM21  
Uncertainty table

 DC resistance standards and sources, low values. Resistance standards, 1   Absolute expanded uncertainty (k = 2, level of confidence 95%) in : 20E06R, with resistance R   Substitution with a DMM   Temperature: (23 ± 3) °C   Approved on 19 May 2015   Internal NMI service identifier: GEOSTM/EM13 
 DC resistance standards and sources, low values. Resistance standards, resistance boxes, > 0.01 to 1   Absolute expanded uncertainty (k = 2, level of confidence 95%) in : (15E06R + 10E06) to 20E06R, with resistance R   Direct measurement with a DMM   Temperature: (23 ± 3) °C   Approved on 19 May 2015   Internal NMI service identifier: GEOSTM/EM13 and EM15  
Uncertainty table

 DC voltage meters, intermediate values. DC voltage calibrator, > 0.001 V to 1100 V   Absolute expanded uncertainty (k = 2, level of confidence 95%) in V: (2E06U + 3E06) to (6E06U + 600E06), with voltage U   Direct measurement   Temperature: (23 ± 3) °C   Approved on 19 May 2015   Internal NMI service identifier: GEOSTM/EM8 to EM12  
Uncertainty table

